1500mAh Endo Motor Portable Dental Equipment Elitepad Apex Locator
Contact Person : Nature Xu
Phone Number : +8613760266213
WhatsApp : +8613760266213
|Packaging Details :||1 set / pack||Delivery Time :||depending on the quantity and sizes|
|Payment Terms :||T/T, Western Union,Paypal|
|Place of Origin:||Shenzhen , China||Brand Name:||Rogin / OEM|
|Certification:||CE ISO||Model Number:||BLX-10|
|Quantity:||1 Set / Pack||Target Angle:||21°|
|Tube Voltage:||70kv||Tube Current:||1MA|
|High Voltage Generating Circuit:||High Frequency Switch||Time Setting Range:||0.05-1.2 Seconds|
mobile dental equipment,
dental x ray machine
BLX-10 Portable Wireless Dental X-Ray Unit Low Radiation Dental Mobile X-Ray Machine
a) The rated parameters of x-ray production device
◆Tube voltage: 70kv (fixed)
◆Tube current: 1MA (fixed)
◆High voltage generating circuit: High frequency switch
◆X-ray control method: Microprocessor control
◆Time setting range: 0.05-1.2 seconds
◆Type: bipolar fixed x-ray tube
◆X-ray tube Model: DG-073B-DC
◆The size of x-ray tube focal spot: 0.3*0.3mm
◆ANODE heating unit: 8.5Khu
◆Maximum tube voltage: 75Kv
◆Target angle: 21°
◆Total filtration: 1.6mmAL (Origin filtration: 0.8mmAL, Additional filtration: 1mmAL)
◆X-ray irradiation range: 73mm (roundness)
e) Useage voltage: battery DC16.8V
f) Charger: Input AC100-240V, Output: DC16.8V
Method of usage:
1.Upon receiving,open the box and check the product for possible damage during shipping.
|2. Make sure the fittings on the encasement list are packaged within the box.|
3. Install the blue ray tube to the foreside of the ball head,then tighten it.
|4. Turn on the Power；the pilot lamp will be light up.|
5. Press on the button of power, then LCD display the “READY”. Now the equipment is in standby mode. (If there is abnormal, the LCD will show the “ERROR”)
|6. When the machine under the “READY”, press on the button, then it will show the falshing pictures of teeth. The LCD displays the pictures of sixe parts’ teeth in palate and mandible contains anteriors, canines and molars.|
|7. During using the soft-switching, it can not change the setting procedure and need to use the main power to reset.|
|8. After adjusting the model, press on the start key and begin to take a photo. (Notice: gently press 0.5 sec to turn it on)|
9. Turn the power off when the work is finished,press POWER, making sure the pilot lamp is off.Don't keep the machine long in standby mode.
This operation instruction includes usage and technical content of portable x-ray system. In order to operate the machine and ensure the safety, it is used by the law qualified personnel.
Consider the radiational danger for the patients and the users, please ensure the safety and comply with the content of instructions.
This product is the equipment combined with the high frequency switch DC 70KV tube voltage and 1mA tube current X-ray, which can be taken a photograph in the arm or solid and can be used with digitized sensor and film x-ray system.
c) Product Feature
1. The product is adopted by the micro-processor for processing the high-frequency switching start from little Ripple amount DC high voltage to draw stable x-ray output, thereby obtain high-resolution images.
2. The machine takes advantage of stable high-efficiency DC high voltage and takes a photograph with small amount of X-ray, which reduces the X-ray irradiation dose for the patient and user.
3. Multilayer X-ray shielding, give full consideration to the safety of users.