Payment & Shipping Terms:
|Material:||Stainless Steel||Packing:||1 Set Per Box|
|Head:||Mini ,Standard Or Torque Head||Chuck Type:||Key Chuck Type / Wrench Type|
|Noise:||≤60dB||Fucation:||3 Water Spray / Single Water Spray|
dental lab material,
dental handpiece parts
Mini Head Key Type Handpieces High Speed Dental Handpieces with Single Spray
Quick Detail :
1. Mini ,Standard or Torque Head
2. Cartridge for Handpiece
3. Key Type / Wrench Type ( 4 or 2 holes )
4. Turn Smooth, Low Noise
5. Use Best Bearing, Long Durability
6. Air pressure: 0.2 - 0.24 Mpa
7. Rotation Speed : >=360krpm ( Standard )
>=300krpm ( Torque Head )
8. Chuck Type : Push Button or Wrench
9. Bur applicable : φ 1.59 mm ~ φ 1.6 mm * 21 mm ~ 23 mm (diameter * length)
10. Noise : <=60db
|1. Fucation : 3 water spray / Single water spray|
|2.Bur applicable: diameter 1.59~diameter 1.6mm*21mm ~ 23mm ( diameter * length )|
|3.Water pressure: 2.2~2.7kgf|
|4.Turns speed : 350, 000~400, 000 rpm|
1. The need to use a clean , dry , oil-free air compressors for medical use
2. The air pressure between 2.3-2.8kgf / cm2
3. Place the adapter with quick connector at the bottom of a combination of dental handpieces , and confirm that the card lock fixed
4. Press the pedal , handpiece operation and examine whether the normal operation of the spray head . If everything is normal, you can start using the handpiece .
Handpiece routine maintenance:
Fault: the occurrence of undesirable phenomena water
( 1 ) inadequate water supply .
( 2 ) the amount of water is not uniform.
( 3 ) is not a mist of water .
( 4 ) no water completely .
Solution: Check the water blocked the site
( 1 ) pipeline : the turbines removed to check for the amount of water flowing from the pipeline . If there is no water out, it may be pipes or water treatment units system problems ( bad or wrong operation , there is not enough water pressure ) .
( 2 ) interface : the interface conductors commissioning . Without the right amount of water flowing out , it indicates a problem at the interface ( such as clogged scale water supply systems or interface loose , resulting in insufficient pressure ) .
( 3) The body of the phone : the shaft from the turbine to remove , install interface running on . If there is no outflow amount of water, it is the problem turbines within the body ( water holes or dirt clogging turbines within the water supply system clogging ) .
( 4 ) shaft : If the above three normal outflow of water , it is the axis of the core problem (water nozzle clogging ) .
Contact Person: Ted Zhou